Wireless Radon Soil Probe TSRP3
Similar to TSR4 Wireless and USB Radon Probe plus the Radon Soil Probe is adapted for insertion into a bore hole of minimum diameter 50 mm and for operation in places with increased humidity as are caves and mines. Two probe chambers are filled with replaceable special dehumidifier for decreasing of humidity in measuring chamber if the probe operates in humidity environment. If it operates in humidity environment the dehumidifier needs to be replaced or dried. If the relative humidity of environment is above 90% the change interval of dehumidifier is 4 weeks. The battery will last for up to 1 year ensuring autonomous operation of the probe within the bore. Data is possible to continuously download over wireless communication and mobile Central Unit or once the probe is removed from the bore, the recorded and stored data can be downloaded over USB into PC.
The TSRP3 Radon Soil Probe is designed to continually measure radon volume activity in soil. Soil radon concentration changes may precede certain behavior of Earth’s crust and various elements in the bedrock. For example, long-term monitoring of the soil gas radon concentration can be used to predict seismic activity or earthquakes.
|Average measurement sensitivity||0,06 imp/hod/Bq.m-3|
|Measurement uncertainty||< 25% at 300 Bq/m3 per 1 hour;|
< 6% at 300 Bq/m3 per 24 hour
|Measuring chamber capacity||0,044 dm3|
|Dimension||Ø 50 x 280 mm|
- TSRP3 Technical Specifications & Operation Manual
- User manual for usbRADONview application
- USB Driver usbRADONview from Windows 7
- USB Driver usbRADONview up to Windows Vista
- Program Installation usbRADONview 2.6.4
- TERAview program can be downloaded at Central Unit